The ultimate handy solution for measuring conductivity!
The Bluelab ppm Pen is the latest innovation for hand held conductivity meters. It provides a compact, quick and handy way to manage the success of your crops.
The Bluelab ppm Pen simply tells you that your plants have the right amount of nutrient in your solution and lets you know if adjustments are needed.
Display options available in degrees Celsius, degrees Fahrenheit, EC, CF, ppm 500 and ppm 700.
Teamed with our Bluelab pH Pen, you have a handy, convenient and accurate system for ensuring optimum plant growth and health.
- Measures conductivity and temperature
- Selectable units for conductivity and temperature
- Backlit LCD display
- Calibration optional
- Successful calibration indicator
- Hold reading function
- Low battery indicator
- Fully waterproof
- Auto off function
- Automatic temperature compensation
- 1 x AAA alkaline battery included
- 1 year guarantee for Bluelab ppm Pen
- Tech Specs
Brand Bluelab SKU HGC716482 Weight (lb.) 0.3 Height (in.) 1.3 Prop 65 No Lead Time Ships in 7 Days Length (in.) 8.2 Width (in.) 2.2 Warranty 1 Year UPC 9421024920227 UL Listed No
Bluelab Conductivity Pen Videos►
- Product Q&A
Bluelab Conductivity Pen Questions & Answers
My nutrient readings are inaccurate - why?
- The probe could be contaminated
- The temperature of the solution could be very different to the temperature of the probe
- Clean the conductivity probe as described in the related article
- Wait 5 - 10 minutes for the reading to stabilize
What ppm scale should I use?
ppm measures parts per million.
ppm is known as dimensionless quantities; that is, they are pure numbers with no associated units of measurement. A mass concentration of 2mg/kg - 2 parts per million - 2ppm - 2 x 10-6.
There are many different scales used for different industries around the world and for many different reasons! Did you even know there are more than two scales? The most widely used scales in Hydroponics are the 500 scale, 650 scale and the 700 scale.
CF and EC are measures of electrically charged nutrient ions in a solution.
Pure water will not conduct electricity. Water usually conducts electricity because it is full of impurities, in our case, electrically charged nutrient ions. The two black dots on the end of Bluelab nutrient probes are called electrodes. When these are placed in a solution, an electrical current passes from one electrode, through the water to the other electrode and counts the number of electrically charged ions present. This represents the units measured - CF or EC.
What's the difference?
The ppm 700 scale is based on measuring the KCl or potassium chloride content of a solution. The ppm 500 is based on measuring the NaCl or sodium chloride content of a solution and is also referred to as TDS - total dissolved solids. Individual nutrient ions have different electrical effects! The true ppm of a solution can only be determined by a chemical analysis. ppm cannot be accurately measured by a CF or EC meter. They are present on Bluelab products as a conversion guide only. The conversion is as follows:
- 2.4EC x 500 = 1200ppm (500 scale) or 1200ppm / 500 = 2.4EC
- 2.4EC x 700 = 1680ppm (700 scale) or 1680ppm / 700 = 2.4EC
If you are reading from a book that says you should grow your crop at 1100ppm - how do you know which scale the writer is referring to? Is the scale on your ppm meter right for the job? If the book was written in the USA, it could be the 650 or 500 scale. If the book is written in the UK, it could be the 700 scale. If it was written in Australia, well it could be any of the three!
If you must grow using ppm, you will need to know the following:
- What ppm scale is the book referring to?
- What ppm scale is your meter using?
- Which standard or calibration solution should you use for your meter?
- What ppm scale is the nutrient formula referring to?
Customer Reviews 3 item(s)
- calibrates with the solution
- if it's reading on par remains to be seen.
- Bluelab is a must have
- I have tried a few other brands (oakton, Hanna etc.) and have had them fail or just not read properly in some situations. I'd recommend to just go with bluelab from the start. Save yourself the hassle!
- Satisfied but not impressed...
- I've always thought of blue lab products as quality (still do) but this ppm/ec pen doesn't read anything below 100ppm, so checking the ppm of your water isn't possible for whatever reason... when I purchased mine it stated it read from 0 to ???? but it doesn't read anything below 100 (as I said) and the last digit is always a 0.. say your solution is 1255ppm, it'll read 1260... Obviously 5ppm isn't the end of the world but when you wanna know what your reverse osmosis unit is putting out, this meter isn't gonna help you...