The GroLine storage solution is prepared with reagent grade chemicals that can be used to ensure optimum performance of your pH and ORP electrodes. It is necessary to store a pH electrode in a solution in order to keep the glass membrane of the pH electrode hydrated. Ideally, a storage solution should be used; never store an electrode in distilled or deionized water. The GroLine storage solution is specifically formulated to minimize microbial growth and to prevent any diffusion/osmotic effects from storing a probe in a solution with the highly concentrated inner reference electrolyte. Storing your pH and/or ORP electrodes in a storage solution will also keep the junction clear. Maintaining the hydrated layer and clear junction will help to ensure a fast and reliable measurement.
The GroLine storage solution can be used to store your pH electrode. To ensure an optimum response time, the glass sensor tip and the reference junction of the pH electrode should be kept moist and not be allowed to dry out when not in use. Placing the electrode in a small glass filled with storage solution or replacing the solution in the protective cap is a suitable way to store the electrode. The GroLine storage solution should also be used to rehydrate the electrode after a cleaning procedure by soaking for at least one hour before taking measurements. In the case of cleaning your pH electrode in one of our cleaning solutions, we recommend soaking the electrode in the GroLine storage solution for at least one hour before taking measurements.
GroLine Storage Solution Features
NOTE: Never store pH or ORP electrodes in distilled or deionized water. Doing so will shorten the life of the sensor.
Sign-up and save! Get exclusive discounts & the latest product releases.
Expert Advice Toll-Free
© 2011-2022 Hydrobuilder.com - 312 Otterson Dr STE D , Chico, CA 95928 USA
All trademarks and trade names are the property of their respective trademark holders.