The most common reasons for pH meter error is dry storage of the electrode and electrode contamination. These cause loss of accuracy, reduced performance and slower response times. Thoroughly clean the electrode after use and always store your pH meter with the cap firmly in place. Frequency of meter calibration is crucial and will optimize your pH meterÕs performance. For calibration of pH Meters at pH 7.0, pH Buffer 7.0 @ 25¡ C (77¡ F) based on international (APHA) standard formula.
|Lead Time||Due to social distancing requirements, this item ships in 7-10 business days|
Sign-up and save! Get exclusive discounts & the latest product releases.
Expert Advice Toll-Free
© 2011-2021 Hydrobuilder.com - 312 Otterson Dr STE D, Chico, CA 95928 USA
All trademarks and trade names are the property of their respective trademark holders.